I do not have the instrument and you did not post a critical subVI. Message 24 of I need to work on this,. Please show your latest VI and typical results for the critical range of Here I m attaching my code In which Input voltage block is there there we have to write command for input voltage i.
|Date Added:||19 January 2007|
|File Size:||56.18 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I forgot the e X ecute command I know its difficult without labview 5. What do you think the start and stop controls are for? So i want to finish my work with 5.
What are you actually measuring in the first frame and how is it different in the third frame? Please show your latest VI and typical results for the critical range of The sweep operation is a simple mathematics operation and you can also look to see if you have the Ramp Pattern function on the Signal Generation Palette.
For increment this input voltage I tried all the way, But here in this both character term and integer term is there, so its not taking increment step. Message 29 of Message 23 of Here I m attaching my code In which Input voltage block is there there we have to write command for input voltage i.
Message 7 of You can set breakpoints and single step through the code to see what the instrument is actually being programmed for and whether it is triggering or not.
This will pause the program. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 2 of I m facing increment problem only. U just simplify my vi which i sent before, Only problem is 6617 m not at all getting readings between -1 to 1v.
Are you using the VIs which I found here, at ni. Make change and forward it to me.
You can then manually operate the instrument and try to take a measurement. The process to achieve this is loop through this code, but changing the voltage range and hence reading back different data. I just want to know how to give increment in my code.
Message 10 of But problem is each time I’ve to mention. Still now my code is correct only problem, commands how to give, its giving data for 1 volt but not the middle values, for that which command should I use, I tried all the necessary commands in my code. You should post the code you aer using now. What’s the point of the measurement in the first frame?
How to add decimal point in out put data – Page 3 – Discussion Forums – National Instruments
We are getting reading from meter. I m sending what I m using here, so labviea u try for increment in my code. Message 1 of If you don’t get the correct values out, then you may be programming the instrument incorrectly. Message 8 of Sorry sir, this is only subvi.
Labview VI’s for keithley meter – Discussion Forums – National Instruments
Once you determine how the instrument needs to be changed, you will have to determine how to set it up programatically. Determine what needs to be done to the instrument in order to take the measurement.
Message 30 of